( A), and ( A, flexible), HIOKI also provides the (5 A) sensor .. A display screen and operation keys identical to those for the appear, allowing full control of . Measurement time control: Manual/Specified time. POWER QUALITY. INSTRUCTION MANUAL. ANALYZER HIOKI. MEMORY HiCORDER. Trigger input terminal. Anomaly. The Hioki PW is a 3P4W power quality analyzer compliant to Class A standards for diagnosing and recording complex power quality problems in the power.
|Published (Last):||24 July 2017|
|PDF File Size:||10.36 Mb|
|ePub File Size:||7.82 Mb|
|Price:||Free* [*Free Regsitration Required]|
Page 27 Changing the flagging Signaling Incl.
Unbalance Settings2 Unbalance Set unbalance value. Table of Contents Add to my manuals Add. Setting value V Vari. Settings For The Measurement System The setting contents are similar to those hioi in the normal mode but the settings conducted here will take priority when EN mode is selected. To print the manual completely, please, download it.
Got it, continue to print. Preparation for Measurement 2. Example When the start key is pressed at By initializing the settings before starting measurement, EN measurement can be started without making further complicated settings. ON Voltage interruption CH1,2,3: When EN mode is selected, the measurement function userr EN is added to the normal mode, and some functions in nor- mal mode will be limited.
You may be unable to read from or save data to such cards. As such, setting in the system screen of normal mode will not be possible. An optional current sensor is necessary to measure current or power parameters.
The time and level cannot be changed.
Download Program for POWER QUALITY ANALYZER 3196
Page 29 Changing the display method Events example Select from pull- down menu Manul Cancel Display the data from measurement start to the present or end time Display the data for a designated period Previous recorded prior to the present or end time Display the data for a designated period Specific from a specified date Voltage waveform peak, Current waveform peak Changing the flagging Signaling Incl. En Measurement Specifications 5.
When “Previous” is chosen as the display method, “Before: Page 49 Event settings The following items are fixed values and cannot be changed. Short interruption time Settings1 Short int T The time for a short interruption is specified. Compatibility and performance are not guaranteed for SD cards made by other manufacturers. Others No results were found.
POWER QUALITY ANALYZER PW – Hioki
Straight Ethernet cable, supplied with straight to cross conversion adapter, 5 m U range setting Settings1 U Range Set ch range only. As per IEC 8. Don’t show me this message again. Changing the display method Events example Select from pull- down menu Confirm Cancel Display the data from measurement start to the present or end time Display the data for a designated period Previous recorded prior to the present or end time Display the data for a designated period Specific from a specified date Future release By specifying a directory to load the data, the will load the display data for EN along with the time plot data and event data, which are normal measurement data.
Settings1 Settings for the Measurement system.
Limitations In En Mode 2. The details of each screen are explained as follows. Common Display Areas 3. Setting value An interruption up to this period is identified as a “short interruption. OFF Total voltage harmonic Setting function specifications When EN mode is selected, some functions in normal mode will be limited.
Set the values in normal mode as required. Note that only periods shorter than the recorded period can be selected.